Recall 2018-162: Trividia Health- TRUEplus Insulin Syringes - NYC - Consumer
URGENT!!! DRUG RECALL!!! URGENT!!!
|Description ||Lot # || Exp Date || NDC || UPC|
TRUEPLUS SYR .3CC 29G
|NP18196 ||07/22/2021 ||56151171101 || 2129200497|
|TRUEPLUS SYR .3CC 31G CT100 ||NP18130 || 06/25/2021 ||56151173101 || 2129200503|
Trividia is voluntarily recalling the above items/lots due to a defect detected in which a small crack in the top end of the barrel near the needle creates the inability to aspirate insulin into the syringe barrel from the insulin vial. This recall is to the Consumer level. Affected product started shipping April 30, 2018.
Consumers should check their syringe boxes to determine if they have impacted product.
If they have impacted product they should contact Trividia Health Customer Care:
1-800-588-1685 Monday - Friday 8am - 8pm EST or via email: trividia0818CC@trividiahealth.com for help with return and replacement information.
This recall is being conducted with the knowledge of the FDA.
Customer Care: Trividia Health 800-588-1685